Caution: Federal law restricts this device to sale by or on the order of a physician or a practitioner trained and certified in its use. Assisted Hatchling, Embryo Biopsy and PGD are not recommended for routine use in all IVF patients.
Please Note: For animal research and human stem cell applications, the XYClone® should be used.
View PDF Spec Sheet
View PDF Letter of Clearance for LAH (ZILOS-tk)
View PDF Letter of Clearance for Embryo Biopsy
View CDRH Consumer Information
Type: 1460 nm, Infrared Solid State Diode
Power: 300 mW @ focus (Class I)
Laser Pulse Length: Adjustable from 1 microsecond to 3000 microseconds, Pulse slider resolution - 1, 10, and 100 microsecond intervals
Laser Power: Adjustable from 1% to 100%
Laser Mode: Clinical Mode: High, Medium, Low
Firing: By mouse or remote firing by foot control
Aiming Marker: Circle or arrow, adjustable Blink Time after firing. Isotherm Rings, showing peak temps and hole size.
Crosshairs: Activation, size and color set by user (used for positioning embryo and moving automatic stage)
Laser Alignment: Aligned and locked in place at factory. No on-site laseer alignment required.
Target Alignment: Performed interactively on-screen
Standard: 40X, L.W.D. 0.60 N.A., I.R. (High transmission in both the visible and near-infrared, long working distance)
Scale Calibration: Calibrate and store multiple objectives; Calibration performed interactively on-screen
Camera: High Resolution analog or digital color
Screen Resolution: Desktop: 1600 x 1200. Laptop: 1280 x 1024
Image Area: Analog: 640 x 480pixels. Digital: 1392 x 1040
Zoom: 2x, 4x and 8x, with image panning (user defined increments); Laser may be fired under zoom.
Class I Laser: Laser pulse and energy cannot exceed Class I limits.
Energy: Once energy level is selected in Clinical mode, other energy levels are deactivated.
Capture and store unlimted images. Images may be stored in JPG, BMP, or TIF format. Manual and automtaic imge file naming. Image Toolbox included (add text, shapes, and measurements to any captured image)
Double Validation Panel
Set two separate power and pulse length combinations
Image Capture and Storage Utility
Capture and store unlimited images. Images stored in user-selected JPG, BMP, or TIF format. Capture unlimited thumbnail images and select which to save. Automatic image capture on laser fire. Manual or automatic file naming. Images may be saved with graphic overlay.
Real-time Video Capture
Record and store video in .avi format. Ability to set maximum recording length. Automatic and manual file naming options.
Time-lapse Video Capture
Record and store video in .avi format. Ability to define capture interval and set maximum recording length. Automatic and manual file naming options.
Tools allow measurement of various embryo parameters on captured images. Each measure visible as graphic overlay, including length in micron
Zona: 5 zona thickness measurements; Mean & Standard Deviation
Embryo: 2 diameter measurements; Mean & Standard Deviation; Blastomere count.
Pronuclei: 2 diameter measurements for two separate pronuclei; Mean for each pronuclei
Drill: 5 hole size measurements; Mean & Standard Deviation
Ruler: 5 user-defined measurements Mean & Standard Deviation
Reports and Output
Data Input: Patient Data from keyboard; Data from measurements; Input critical fields from ASCII file
Patient Record: Patient information, treatment identifiers, before and after treatment images, embryo evaluation data
Output: Printout of Patient Record; Patient Record stored in JPG format; ASCII output of all numerical and alphanumeric fields in TXT and MER formats (IDS, Recdate compatible)
Add text, ellipses, rectangles, lines to captured images. Measue any area of image with ruler tool. Custom select font, color and line widths.
All brands/models of inverted microscopes (no laser module customization needed)
Dimensions H x W x D in. (mm) / lb (kg)
ZILOS-tk Module: 1.65 (42) x 1.1 (28) x 2.1 (54) / 0.5 (0.22)
Controller: 3.8 (95) x 2.5 (63.5) x 1.8 (46) / 1.0 (0.45)
Desktop Computer: 16.1 (408) x 7.4 (187) x 17 (433) / 26.5 (12)
Monitor: 13.5 (343) x 17.5 (444) x 1.5 (38.1) / 17.0 (7.7)
Laptop Computer: 1.3 (3.3) x 15.3 (388) x 9.9 (251) / 6.2 (2.8)
Input Voltage: Desktop: 100/240 VAC, Monitor: 110-240 VAC, Laptop: 100-240 VAC
Power: Desktop: 300 W, Monitor: 30 W, Laptop: 90W
Line Frequency: Desktop: 50/60 Hz, Monitor: 50/60 Hz, Laptop: 50/60 Hz
|ZILOS-tk Computer Specifications (subject to change without notice)
||Windows 7 Professional
||Windows 7 Professional
||Pentium Dual Core 3 GHz or Core i5
||Intel Core i5
||2 GB DDR
250 GB HD
16x DVD+/-RW SATA
|250 GB HD
||20" flat panel
||15" (typical), XGA
||Serial, Parallel, USB 2.0, PS/2
||Parallel, USB 2.0, VGA, 1394, HDMI, Memory card reader
||10/100 Mps Ethernet
||10/100 Mps Ethernet
Wireless Network (802.11bg)